Question to Sterilizers.com chief trouble shooter:
What might remove burnt char from bottom of a Tuttnauer. Green fluid leaked after cleaning and burnt onto bottom of chamber. Chamber Brite not doing trick.
If there are melted plastic, chamber-brite will not do much. Make sure the dry cycle is not used. This may take a few cleanings.
It can be hard to clean unless you use an acid.
They will have to run another cycle or two using chamber-brite.
1. They can run a ½ package for no more than 30 minutes @250-260 degrees.
2. Vent chamber and empty reservoir, refill with fresh water.
3. While chamber is still slightly warm, use fine scotch-brite or a course side of a dish washing sponge and rub left to
right in the chamber (with the grain)
4. Wipe out chamber
5. Run another clean cycle
6. Do #2 again
7. Do #3 again
8. Do # 4 again
Steam sterilization is a popular choice for a reason. It is a dependable, non-toxic, and affordable solution that will meet your sterilization needs. However, according to the CDC, steam sterilization can have deleterious effects on some materials. In our article “Why You Need to Contact a Sterilizer Equipment Expert“, we proved to you that we are the premier sterilizer equipment experts. In this article, we will answer the question, “Should I use a steam sterilizer?” (more…)
Here are some additional bullet points:
- Electronic controls and LCD display
- Digital data printer and recorder
- Programable cycles and parameters
- Programable exhaust either fast or slow
Place your order now so you can be on the top of the list of early deliveries
Click here to place your order now
A door of an autoclave recently (October 31st 2013) blew off the hinges, and killed a woman who wasn’t too far from the autoclave. This accident didn’t happen in a dental, medical, veterinary setting, but it could easily happen. Please make sure you maintain your autoclave regularly. I will post later all of the sterilizer manufacturers recommendations as to how often the sterilizer each one of their sterilizer needs to be checked.
Click here Autoclave blew off the hinges to view the report in WBS-TV Atlanta Ch 2
If you work in the medical, dental, veterinary, or podiatry field, there’s a good chance that you need a sterilizer. We know that not just any sterilizer will do, so contact a sterilizer equipment expert to get free information about autoclaves and sterilizers.
If you determine that it is important for you to make sure that your Market Forge STME or STME-L sterilizer is working properly in your laboratory, then you have three (3) options:
1. Temperature Recorder:
This pressure sensitive recorder is specifically
made for use with the Market Forge Sterilmatic
Sterilizer (autoclave). Its use is for monitoring
and recording the sterilizers operating temperature
throughout its sterilizing cycles. One recording
chart can be used for a full 24 hour period.
This is a regular thermometer that can withstand the heat as high as 500f. You place it inside the chamber, run your cycle, and when it is done, you can see the highest temperature that the chamber achieved. Very reliable, and excellent as a proof that your temperature gauge shows the right temperature.
3. Spore testing:
a. Bowie-Dick test card to monitor pre-vacuum steam sterilizer to detect air leaks or inadequate steam penetration.
b. Spore testing to be done away form your location
c. Do it yourself kit
The exhaust condenser is used for condensing exhausted steam from the sterilizer when unit cannot be vented to the outside.
With the sterilizer exhaust condenser kit you can use your sterilizer anywhere without having to worry about direct venting access.
If the answer is yes, be aware.
Be aware of the fact that the majority of the autoclaves that you will find have been around for a while, and most likely, the chamber might cracked.
It takes time for these chambers to develop a crack. But when they do, the process from the time it begins to loose pressure until the time you are ready to throw the towel from frustration can be months or even a year or two, but the chamber eventually will need to be replaced.
If you would like to know more what needs to be done to avoid any issue now, and not to spend lots of money, call me for free consultation. Or better yet, fill out this form, and I’ll get back to you as soon as possible.
If your sterilizer is broken, or doesn’t heat up, or pressure is building up way above the 28psi, or the sterilizer won’t pass a spore test, and … you need help … we can help you. You may click on Local Sterilizer Repairs Near You.
A word of advise. When you eMail or call, please be ready with answers to the following questions: